|Description||A Cover-Slip should be placed on top of the O2k-Stopper to minimize contamination and evaporation of liquid extruding from the capillary of the stopper. The Cover-Slips do not exert any direct effect on oxygen backdiffusion into the O2k-Chamber. Use the Cover-Slip\black (Product ID 24411-01) for optical measurements in the O2k.|
|Link||MiPNet22.11, Previous Product ID: 24410-01|
- Cover-Slips cannot be placed on top of the stoppers in O2k-MultiSensor applications with additional sensors inserted through the stopper, or while TIP2k needles are inserted into the O2k-Chamber.